TM-2430 PC2 Long-term Sphygmomanometer
With the 24-hour blood-pressure gauge Boso-TM-2430 PC2, you decided on a blood-pressure computer of the latest technology.
The device was designed specifically for 24-hour measurements, and works on the oscillometric measuring principle. Here, the pressure-fluctuations caused by pulse-waves and transmitted by the cuff (oscillations), are stored and evaluated by the micro-processor.
The big benefit of this measuring method is that no microphone is necessary, on whose exact positioning the accuracy of the measurements is largely dependent.
This blood-pressure gauge conforms to the European norms, based on the Medicine Product Laws, as well as Europanorm 1060, Part 1: "Non-invasive blood-pressure gauges - Common requirements" and Part 3: "Supplemental Requirements for electro-mechanical blood-pressure measuring-systems". The measuring-technical control - at least every two years - can be performed either through the manufacturer or through automated maintenance services, in accordance with the Medical Product-Operator Regulations.
- network-compatible with GDT-port
- small recorder 100 x 80 x 27 mm, 220 g
- with many accessories
- 24-hour blood-pressure gauge TM-2430 PC2
- two battery-packs with three batteries each
- cuff for adults (standard/washable)
- hip-pocket with removable carrying strap and belt
- user instructions for: TM-2430 PC2
- CD-ROM Boso profile-manager
|Product:||Blood-pressure gauge for 24-hour measuring|
|Nominal voltage:||3 x 1,2 V DC|
|Power supply:||3x NiMh-re-chargeable batteries (Mignon)|
|Classification:||Defibrillation-protected device type BF|
|Measuring range:||40 - 280 mmHg 30 - 200 pulse/min|
|Accuracy:||Pressure-display ±3 mmHg, pulse ±5 %|
|Operating conditions:||a+10 °C to +40 °C, 20-85 % rel. humidity|
|Storage conditions:||-20 °C to +70 °C, 20 - 85 % rel. humidity|
|Measurements (W x H x D):||72 mm x 27 mm x 100 mm|
|Type of device:||humeral blood pressure monitor|
Customer assessment ⦁ TM-2430 PC2 Long-term Sphygmomanometer
Sehr gutes Preis-Leistungs-Verhältnis, in dieser Preisklasse gibt es meines Erachtens kein besseres Gerät, dazu der gute und Schnelle Service der Firma BOSO. Ist bereits unser 2. Gerät,...